The TB4MM, manufactured by HS Hyquest Solutions, is a high-end tipping-bucket rain gage that monitors rainfall in metric rather than US units. It has a 20.3 cm orifice and measures rainfall in 0.2 mm increments. The TB4MM is ideal for locations where intense rainfall events may occur. This tipping bucket is compatible with all Campbell Scientific dataloggers, and it is used in environmental monitoring applications.Read More
The TB4MM funnels precipitation into a bucket mechanism that tips when filled to its calibrated level. Each tip is marked by a dual reed switch closure that is recorded by a datalogger pulse count channel. After measurement, the water drains through two orifices (accepts 12 mm tubing) in the base, allowing the measured water to be collected in a separate container.
The TB4MM contains an internal siphon mechanism that causes rain to flow at a steady rate to the tipping bucket mechanism (regardless of intensity). The siphon allows the sensor to make accurate measurements over a range of 0 to 50 cm per hour.
|Sensor Type||Tipping bucket with siphon and dual reed switch|
|Rainfall per Tip||0.2 mm (0.008 in.)|
|Measurement Range||0 to 700 mm/h (0 to 27.6 in./h)|
|Accuracy||700 mm/h (27.6 in./h) maximum rate per hour|
|Operating Temperature Range||0° to 70°C|
|Humidity Range||0 to 100%|
|Cable Type||Two-conductor shielded|
|Drain Tube Size||Both filters accept 12 mm (0.47 in.) ID tubing.|
|Orifice Diameter||20 cm (7.9 in.)|
|Height||33 cm (13 in.)|
|Weight||2 kg (4.4 lb) with 7.62 m (25 ft) signal cable|
Note: The following shows notable compatibility information. It is not a comprehensive list of all compatible or incompatible products.
The base of the gage is supported by three legs. A CM240 Leveling Base and Mount or a user-supplied baseplate with leveling capability is recommended. The CM240 may be attached to a CM300-series mounting pole, or to a user-supplied 1.5 IPS (1.9-inch outer diameter, unthreaded) pipe (see Ordering Information). A concrete pad is recommended.
Campbell Scientific offers the 260-953 Wind Screen to help minimize the affect of wind on the rain measurements. This wind screen consists of 32 leaves that hang freely and swing as the wind moves past them.
|CR1000||CR3000||CR9000X||CR6 Series||CR300 Series||CR1000X|
Number of FAQs related to TB4MM-L: 12
Expand AllCollapse All
The following are some suggestions to try:
Not registering any tips, missing tips, or double tips.
If data is not being logged correctly, there are two likely causes:
Not enough tips or too many tips.
The 260-953 Alter-type rain gage wind screen consists of 32 heavy metal leaves that hang freely and swing as the wind moves past them. The swinging leaves act as a wind damper and help minimize the effect of wind on the rain measurements without adding additional turbulence.
The most common errors are either that the rain gage appears to have drifted out of calibration or that the tips are not being correctly recorded by the data logger.
When a rain gage is out of calibration, it is usually because of the buildup of dirt and grime on the internal surfaces of the tipping bucket mechanism. Cleaning the internal surfaces usually brings the rain gage back into calibration. It is also possible that a rain gage is out of calibration because it is no longer level.
To minimize the possible occurrence of calibration errors, perform routine cleaning and maintenance of the rain gage at least once every three months. The environmental conditions at a particular site may require a facility to perform cleaning, leveling, and maintenance on a much more frequent schedule.
Most Campbell Scientific sensors are available as an –L, which indicates a user-specified cable length. If a sensor is listed as an –LX model (where “X” is some other character), that sensor’s cable has a user-specified length, but it terminates with a specific connector for a unique system:
If a sensor does not have an –L or other –LX designation after the main model number, the sensor has a set cable length. The cable length is listed at the end of the Description field in the product’s Ordering information. For example, the 034B-ET model has a description of “Met One Wind Set for ET Station, 67 inch Cable.” Products with a set cable length terminate, as a default, with pigtails.
If a cable terminates with a special connector for a unique system, the end of the model number designates which system. For example, the 034B-ET model designates the sensor as a 034B for an ET107 system.
Not every sensor has different cable termination options. The options available for a particular sensor can be checked by looking in two places in the Ordering information area of the sensor product page:
If a sensor is offered in an –ET, –ETM, –LC, –LQ, or –QD version, that option’s availability is reflected in the sensor model number. For example, the 034B is offered as the 034B-ET, 034B-ETM, 034B-LC, 034B-LQ, and 034B-QD.
All of the other cable termination options, if available, are listed on the Ordering information area of the sensor product page under “Cable Termination Options.” For example, the 034B-L Wind Set is offered with the –CWS, –PT, and –PW options, as shown in the Ordering information area of the 034B-L product page.
Note: As newer products are added to our inventory, typically, we will list multiple cable termination options under a single sensor model rather than creating multiple model numbers. For example, the HC2S3-L has a –C cable termination option for connecting it to a CS110 instead of offering an HC2S3-LC model.